Weiming Reported Si Geometry Problem

• Weiming Y reported that he was seeing shifts in residuals on L3 and L5 not related to Lorentz angle.

• Also claims that readout order on shallow stereo side of L6 seems reversed!

• Use v4.0.0int2 to look at

– 2 GeV “good” Si tracks – Find nearest cluster to track intersection at wafer

Shallow Stereo Weirdness

• There was an unexplained 1 strip offset for L3 and L5 seen in Commissioning Run data of Oct 2000.

• As of November 2, 2000 this unexplained shift was put in by hand

More Shallow Stereo Weirdness

• The Lorentz angle correction for ISL is not yet in (Patrizia?)

• But we should still see something more or less ok on shallow stereo side of L6.

Remove Artificial Strip Shift

• Removing strip shift from L5 shallow stereo readout works

Remove Artificial Strip Shift

• Flipping ISL also shallow stereo readout works • But...

Remove Artificial Strip Shift on L3

• Removing strip shift from L3 shallow stereo readout DOES NOT work!